MESR-100 V2 ESR / Low Ohm In Circuit Capacitor Meter 0.001 to 100RMIB-INSTRUMENTS"Can Buy It Now in my eCRATER store"Features:JINGYAN MESR-100 V2Auto-ranging capacitor ESR and Low Ohm MeterMeasuring range from 0.001 to 100.0R, support IN CIRCUIT Testing.Using true 100 KHz sine wave to measure the ESR value, which is equal to the testing method of capacitor manufacturer.In the market, there is some technique using short pulse method to testing, but the value will be varying vs the capacitance and sometimes reading is different from the manufacturer's value.What is ESR of capacitor?There is a series resistor inside capacitor, using 100kHz to remove the impedance 1/(2*pi*F*C), the impedance become small, and we can then measure the true series resistor value.A bad E-capacitor will have larger ESR and create large ripple rather than filtering noise. Normally, a big capacitor is larger than 3 ohm.Using this theory, we can measure the capacitor is bad/ damaged or good in condition.Because our ESR meter only applies less than 15mV DC or peak to peak on a good capacitor, as a result we can use it as in circuit test. Because this low testing voltage, it cannot turn on the semiconductor inside a circuit under testing.During repairing TV, LCD, Audio board, etc. we can in circuit testing the capacitor is good or not.*Dual terminal, for fast and easy inspect general capacitor or resistor, a printed ESR table for fast check Compare MESR100 old V1 and new V2 Improvement: 1)Change square wave to sine wave 100 KHz, reduce square wave’s high frequency component, and affect the reading passing the test leads and capacitor.2)Higher Resolution up to 0.001 ohm.3)128X64 dot matrix LCD, with more larger value display and information4)Embedded 25V capacitor table at LCD, auto display the capacitor is good or bad reference to common 25V electrolytic capacitor.5)New plastic case, curve design for hand carrying. New stand for 60 degree stand on desk.6)Use 2X AA battery, more convenience and longer battery life than 9V battery.7)Support external USB power, using standard micro-USB port.Specification:RangeAccuracy (After Zero, tested with 1,10,100R resistor)Refresh Time (Manual Mode)*Auto mode take 0 to 2s more time depend on value0.000 to 1.000R1%+2Digit~0.4s1.000 to 10.00R1%+1Digit~0.4s10.00 to 100.0R2%+1Digit~0.4s*Accuracy maybe affected by the test lead’s length and contact point resistance.Standard test leads is 15cm, for longer test leads need special shielding to affect the EMI inference (Can contact sales’ person for more detail)1)Accuracy: Up to 1% (detail on above table)WideMeasurement Range: >1uF (for 0.1uF the error will be larger on equation1/(2*pi*F*C) @ 100Khz )2)HighResolution:4 digit, or 0.001Ohm @ 1 Ohm range3)Measuring voltage: <~40mV RMS (TESTING VOLTAGE)4)Clamping voltage: ~0.15V (open voltage)5)Battery 2X AA 1.5V battery6)External Power: 5V micro USB 7)Operating current 0.02A8)Battery Life time: >80 HoursOperating Introduction:1)POWER ONPress and hold the ON/ OFF circle orange button for 2 to 3 second to power on.Press and release the ON/ OFF button to shut down meter.2) AUTO/MANUAL mode:i) AutoRange:Press andrelease the RANGE button and at the first line of LCDwill show “AUTO:"At auto mode meter will automatically select the best range todetect.ii) Manual Range:Scroll the manual range from 1R, 10R and 100R range by press and release the RANGE buttonIn LCD will show MANUAL at first LCD line and show at second line: 0-1R, 1-10R and 10-100R.3) 1 KeyZero:Short circuit the test leads’ terminal, cancelling the wire resistance.Press and release the “ZERO" button, LCD shows“ZERO" and wait the zero disappear.If you are using the meter array socket, you need to use a short pin to short circuit to set zero.4) Backlight:LCD backlight will on during power on5) Auto Sleep:Around 10 hours for not testing, it will shut down automatically to save power.6) OVERFLOW or OL:Display overflow when the value is out of range, you can check that you are zero correctly.7) ESR TABLE:It is for reference only, different manufacturer of capacitor has different ESR, better to test a good cap ESR and compare with what you are testing, usually bad cap with higher ESR few times larger than good one.8) SOFTWARE ESR:Put the 25V Electrolytic capacitor value to LCD, and compare, GOOD if C < XXuF, means the capacitor maybe good condition if it is less than XXuF9) PLASTIC STAND:If you don’t use the stand, press and push the lock position:Package Content1 unit MESR-100 V21 pair 15cm test clips (as photo)For longer test leads you can select in our store* For measuring capacitance, please purchase M6013 capacitance meter in our store" http://cgi.eCRATER.com/ws/eCRATERISAPI.dll?ViewItem&rd=1&item=320799847440MIB-INSTRUMENTSPaymentShippingWe accept Paypal, credit card through Paypal. Buyer may need to pay for import tax depend on your country's custom. You can ask us for more detail. We ship to international and item will be shipped by airmail from HongKong. Items are shipped within 2-3 business day after receive cleared payment. An additional USD2 for insurance Most item will arrive within 8-14 business days. It depend on your location. In some case, ship to some countries will take over 3 weeks, especially brazil, Italy and France . Powered by eCRATER . List your items fast and easy and manage your active items. Powered by eCRATER . List your items fast and easy and manage your active items.